Overall Size: 2055 mm L X 900 mm W X Hgt. 560 mm(Approx.)
Suitable for Mattress Size: 1980 mm L X 890 mm W ( Approx.)
Bed frame is made up of 60 mm x 30 mm x 16G CRCA rectangular tubes.
4 sleeves of 40x40x 100 mm long welded at four corners with nylon insert to accommodate the bed bows
Four- flush I.V. Rod locations with Nylon inserts to avoid rusting
Three sections top is made up of 18G CRCA sheet
Two fixed shelves double press bend on four sides and uniformly perforated holes 23 mm diameter
Backrest and leg end section is welded to 25.4 mm diameter x 16G CRC Sheet Backrest section can be operated screw mechanism with folding handle non detachable in SS.
Legs made from 40 mm x 40 mm x 18gauge ERW tube with horizontal tube 25.4 x 25.4 x 18 swg tube.
The end of the legs that fits to the bed welded to a 150 mm long at least 3 mm thick section.
Provided with Nylon shoes with metal inserts.
Detachable Under side shelf at least 220 mm wide provided under for keeping patient belongings. To bed the bed made form 19 mm dia round ERW tube with wire mesh welded to it.
Detachable waste basket of 300 dia aprox fitted on a revolving ring made form 12 mm dia which can be installed to either side of the bed at least 150 mm clear from the ground
Bows removable 31.25 mm diameter 18 swg ERW tube with internal horizontal support in 19 x 19 18 swg ERW tube
Partial panel in Poly Propylene (instead of laminated panels) fitted inside the Bow to cover more than half of the bow
100 mm thick mattress with sections as per the bed with Rexene cover in pre approved color
Finish: All components are pretreated in separate eight tank process for better finish, good adhesion and corrosion protection. Process includes Hot Degreasing, Derusting, Activation, Phosphating& No’s of Water rinses as per IS 3618 - 1966 class ‘C’ type and then pretreated
materials is coated with epoxy powder with film thickness of 60 microns and then oven baked at 180 °C