Overall Size: 1900 mm L X 900 mm W X Hgt. 560 mm(Approx.)
Suitable for Mattress Size: 1980 mm L X 890 mm W ( Approx.)
Bed frame is made up of 60 mm x 30 mm x 16G CRCA rectangular tubes.
4 sleeves of 40 mm x 40 mm x 100 mm long welded at four corners with nylon insert to accommodate the bed bows
Four- flush I.V. Rod locations with Nylon inserts to avoid rusting
The bed having 3 panels along them uniformly perforated with 23 mm diameter holes and welded to the main frame.
Each panel press bent an all 4 side and also having an additional support from beneath to make it sturdy.
Legs made from 40 mm x40 mm x 18gauge ERW tube with horizontal tube 25.4 x 25.4 x 18 swg tube.
The end of the legs that fits to the bed welded to 150 mm long at least 3 mm thick section. Provided with Nylon shoes with metal inserts
Detachable Under side shelf at least 300 mm wide to be provided under for keeping patient belongings to bed the bed made form 19 mm dia round ERW tube with wire mesh welded to it.
Detachable waste basket at least 220mm dia aprox fitted on a revolving ring made form 12 mm dia which can be installed to side of the bed of at least 150 mm clear from the ground
Bows removable 31.25 mm diameter 18 swg ERW tube with internal horizontal support in 19x19mm, 18 swg ERW tube
Partial panel in Poly Propylene (instead of laminated panels) to be fitted inside the Bow to cover more than half of the bow.
100 mm thick mattress with Rexene cover in pre approved color
Finish: All components pre-treated in separate eight tank process for better finish, good adhesion and corrosion protection. Process includes Hot Degreasing, Derusting, Activation, Phosphating & No’s of Water rinses as per IS 3618 - 1966 class ‘C’ type and then pretreated materials is coated with epoxy powder with film thickness more than 60 microns then oven baked at 180°C